[teqc] Ashtech Z12 spurious rollover bug
fmeyer at obs-besancon.fr
Wed Nov 9 05:43:25 MST 2016
A number of Ashtech Z12 receivers had a spurious rollover event
at the week boundary this sunday ; it makes them generate
raw files with bogus (1024 week late) dates,
both in the file name and the content.
It looks as if firmware had somehow determined that 1999 aug 21
is still to come, therefore ceased to add 1024 to WN hence
jumped back to 1997 march 23 (WN 898) instead of 2016/11/06
Proper week number must be supplied as an argument to teqc
at processing time. Data quality looks otherwise unaffected
at that point.
Our sample is :
receiver type: ASHTECH Z-XII3
receiver firmware: 1L01
but other colleagues have been affected, maybe with different
Can this be fixed anyhow ?
François Meyer Tel : (+33) 3 81 66 69 27 Mob : 6 27 28 56 83
Observatoire de Besancon - BP1615 - 25010 Besancon cedex - FRANCE
Institut UTINAM * Universite de Franche-Comte * CNRS UMR 6213 ***
More information about the teqc